Technical parameters of X-ray machine
High pressure generator
1.1.Maximum output power of high voltage generator:50KW.
1.2.Maximum tube voltage:150KV.
1.3.Inverter frequency of high voltage generator:50KHz.
1.4.Maximum tube current:greater than or equal to 600mA.
X ray tube bracket
2.1 column type X-ray bulb holder.
2.2.Vertical movement range:1200-1900mm.
2.3.Longitudinal movement range:1700-2500mm.
2.4 horizontal movement range:greater than 220mm.
Photography flat bed
3.1 horizontal vertical movement range of bed surface:850-1000mm.
3.2 horizontal horizontal movement range of bed surface:220-250mm.
3.3 automatic control of electrical fixation.
3.4.With active filter grid or fixed filter wire,the gate ratio is greater than or equal to 10:1,and the grid density is greater than 40 lines/cm.
Chest X-ray frame
4.1.The grid ratio of the active filter grid or fixed filter line is greater than or equal to 10:1,and the grid density:40-78 lines/cm.
5.Radiological safety protection:comply with international radiological safety protection standards.